"ST60 Instuments may be adapted for use with the a/c/e/gS-Series MFDs through use of a SeaTalk to SeaTalkng Converter Kit.
Any SeaTalk instruments, autopilot components, GPS sensors, and chartplotters featuring a 1st generation SeaTalk interface…"
"Welcome to the Raymarine Technical Forum,
Wind transducers cannot be directly interfaced to Raymarine MFDs. While NMEA 2000 wind transducers may be featured within a SeaTalkng/NMEA 2000 backbone featuring a Raymarine a/c/e/gS-Series MFDs, the…"
My 48" open array SHD Digital is not working. The rest of my E120W system seems to be working correct. I was trying to hook up my Compu-course 220 Autopilot to my system and I was into lots of menu items. Later I noticed my array wasn't parked in the location its supposed to be in. It was off by near 180 degrees. I then started finding out why. My E120W powered up normally but when I went to activate the radar, I noticed the radar buttons not lit and the scanner was, but in "off" tab. I…See More
From a performance standpoint, there is no advantage to either solution. However, you need to follow the following basic guidelines of SeaTalkng networking:
no more than three SeaTalkng devices may daisy chained to form a single…"
I want to add a second P70 in the cockpit. The existing unit is in the nav station. I'm also adding an I70 in the cockpit. Can I daisy chain the two cockpit devices together and run off a separate spur, or is it better to spur off the existing P70?See More
New CP300 supplied by Raymarine replacing a failed DSM300.Existing P319 depth transponder and P120 speed/temp.C90W (software 2.35) multifunction display.After experiencing loss of depth on the first seatrial of the CP300, I did some reading here and saw that the recommended CP300 software for legacy screens (ie. my C90W) was v3.25I downgraded the CP300 software from 4.04 and now have the CP300 reporting itself as a DSM300 (which is normal) and version 3.25. (This took five programming cycles…See More
Your procedure is correct and, yes, you will be able to select the desired source. The existence of an Airmar Weather Station in my network has no effect on this problem. The symptoms are the same with or without the Airmar in the…"