Hello,I have a Raymarine SL70 Pathfinder. Everything is hooked up correctly as I have not had any problems with my system the past few years. However a few weeks ago, one of the copper prongs pull…Continue
I would start by making sure that your cable connections both on the back of the SL70 and inside of the radar array are clean tight and correct. No bent broken or corroded pins / wires, especially with the 8 pin wago connector inside…"
Hello,I have a Raymarine SL70 Pathfinder. Everything is hooked up correctly as I have not had any problems with my system the past few years. However a few weeks ago, one of the copper prongs pull out of the Seatalk socket in the back of the unit. I sent the unit back to Raymarine who then replace the prong. When I test the Seatalk transmission it currently works fineThe problem that I know have is that my Arrary Spins, but I cannot see it detecting anything on my screen (which looks like…See More
"Welcome to the Raymarine Technical Forum Scott,
Please note that the US NOAA Raster and Vector Charts do not require that a LightHouse ID be programmed onto the microSD memory card and as such do not require entry to the Charts…"
I am interested. Can you confirm it is stil functional before it gets dismounted? I should know shortly if the one I found in Oriental, NC is available or not. They are asking $250 for that one, in working conditions, from the thech there -…"
"Welcome to the Raymarine Technical Forum Bob,
Please note that the US NOAA Raster and Vector Charts do not require that a LightHouse ID be programmed onto the microSD memory card. However, if downloading the LighHouse Charts for Canadian…"
"A printed version of the diagram is much more legible than viewing the attachment and it shows the 740 input on the standard rate port connected to the NMEA OUT (+) on the T122. I'll be connecting the 740 GND to the NMEA OUT (-) today."
The dump should only be performed if the problem re-occurs. In fact, it is recommended that the Error Logs be erased (HOME->SET-UP->MAINTENANCE->DIAGNOSTICS->ERASE LOGS) within your system. That way, should the problem…"
In fact the methods provided previously has been used to support interfacing countless Raymarine and third party VHF radios to Raymarine's a-Series MFDs. This specific Uniden VHF radio is one of the very few which have been…"
It is recommended that the chart card be removed from the MFD's reader and inspected for bent, broken, or missing pins. Should none be observed, then the MFD's memory card reader may be tested by attempting to archive…"
Should you have recently purchased older inventory not having a NAG card, then it is recommended that Navionics be contacted to replace/upgrade your chart card. Navionics has corresponded with Raymarine and has created a policy to…"
It is certainly possible to that fault may be data driven (i.e. caused by a defect within the cartography). The MFD's design is such that any time that the MFD encounters a condition causes the MFD's software to become unstable…"
A SeaTalk to SeaTalkng Converter Kit would not be used for this purpose. A SeaTalkng 5-Way Connector would instead be used to interface these products. Please click here to view the Forum discussion in which I had…"